BIOLOGICAL INDICATOR 3M ATTEST 1262, BOX 100 vials
INDICATOR STEAM 3M ATTEST 1262
Attest biological indicator in 1262 (brown cap)
E 'it designed for the control of steam sterilization processes.
The presence of Geobacillus stearothermophilus spores is detected by an evident change in color (the color of the culture broth turns to yellow). The yellow color indicates that the sterilization process was not effective. The final reading of a negative result (the culture broth remains purple color) is carried out after 48 hours of incubation.
Monitoring frequency: The Attest biological indicators should be placed in a tray or in a test pack suitable and should be used to control all loads. Only by doing this is guaranteed improvement of the sterilization procedure.
Warning: In the biological indicator in plastic vial is contained a glass ampoule. • Rupture or excessive handling biological indicator before cooling can cause breakage of the glass ampoule. • Wear protective goggles and gloves when removing the biological indicator from the sterilizer. • Wear safety glasses to break the biological indicator. • The biological indicators must be kept for the cap during the ampoule breaking and the resulting pat. • Do not use your fingers to break the glass vial. • Do not rotate the biological indicator between your fingers to wet the strip of spores from the culture broth
Precautions: Do not use the Attest 1262 biological indicator to check: 1. the steam sterilization cycles with removal of air by gravity to 134 ° C (270 ° F), time = 3 minutes; 2. sterilizers Dry heat, to chemical vapors, with ethylene oxide or other low temperature sterilization processes. 100 vials pack
N.B. the photograph is indicative, the appearance and brand of the product may differ depending on availability.